Prior to handling test materials, performing equipment setups, and/or conducting this method, testers are required to read “SAFETY AND HEALTH ” in Section D of this method. It is the responsibility of whoever uses this method to consult and use departmental safety and health practices and determine the applicability of regulatory limitations before any testing is performed. A. SCOPE This method describes the procedure for the direct calibration of expansion pressure devices, employed in the R-value test, California Test 301, by the use of a countertype mass scale, modified with a hold-down table and platform stud arrangement to accommodate the expansion pressure device
To submit an update or takedown request for this paper, please submit an Update/Correction/Removal Request.