Location of Repository

The European Synchrotron Radiation Facility (ESRF) located in Grenoble, France is a joint facility supported and shared by 19 European countries. It operates the most powerful synchrotron radiation source in Europe. Synchrotron radiation sources address many important questions in modern science and technology. They can be compared to “super microscopes”, revealing invaluable information in numerous fields of diverse research such as physics, medicine, biology, geophysics and archaeology. For the ESRF accelerators and beam lines to work correctly, alignment is of critical importance. Alignment tolerances are typically much less than one millimetre and often in the order of several micrometers over the 844 m ESRF storage ring circumference. To help maintain these tolerances, the ESRF has, and continues to develop calibration techniques for high precision spherical measurement system (SMS) instruments. SMSs are a family of instruments comprising automated total stations (theodolites equipped with distance meters), referred to here as robotic total stations (RTSs); and laser trackers (LTs). The ESRF has a modern distance meter calibration bench (DCB) used for the calibration of SMS electronic distance meters. At the limit of distance meter precision, the only way to improve positional uncertainty in the ESRF alignment is to improve the angle measuring capacity of these instruments. To this end, the horizontal circle comparator (HCC) and the vertical circle comparator (VCC) have been developed. Specifically, the HCC and VCC are used to calibrate the horizontal and vertical circle readings of SMS instruments under their natural working conditions. Combined with the DCB, the HCC and VCC provide a full calibration suite for SMS instruments. This thesis presents their development, functionality and in depth uncertainty evaluation. Several unique challenges are addressed in this work. The first is the development and characterization of the linked encoders configuration (LEC). This system, based on two continuously rotating angle encoders, is designed improve performance by eliminating residual encoder errors. The LEC can measure angle displacements with an estimated uncertainty of at least 0.044 arc seconds. Its uncertainty is presently limited by the instrumentation used to evaluate it. Secondly, in depth investigation has lead to the discovery of previously undocumented error-motion effects in ultra-precision angle calibration. Finally, methods for rigorous characterisation and extraction of rotary table error motions and their uncertainty evaluation using techniques not previously discussed in the literature have been developed

Topics:
TK, QC

OAI identifier:
oai:wrap.warwick.ac.uk:2775

Provided by:
Warwick Research Archives Portal Repository

Downloaded from
http://wrap.warwick.ac.uk/2775/1/WRAP_THESIS_Martin_2009.pdf

- 17123 Optics and optical instruments - Field procedures for testing geodetic and surveying instruments
- (2001). 17123-3:2001(E) Optics and optical instruments - Field procedures for testing geodetic and surveying instruments Part 3: Theodolites.
- 9001: Quality management systems — Requirements. Third edition ed. 2000: International Organization for Standardization.
- (1991). A history of mathematics. 2nd ed.
- (1989). A history of pi (pi).
- (2007). A Modern Calibration Bench: Calibrating Survey Instruments.
- (1997). A Multi-Point Method for Spindle Error Motion Measurement. CIRP,
- A multipoint method for spindle error motion measurement.
- (1990). A new method of theodolite calibration.
- A novel technique for calibration of polygon angles with non-integer subdivision of indexing table.
- (2002). A reply to "Definitions of the units radian, neper, bel and decibel"
- A Solution to Atmospherically Induced Problems in Very High-Accuracy Alignment and Leveling.
- (2004). Acoustic method for determination of the effective temperature and refractive index of air in accurate length interferometry. Optical Engineering,
- (2008). Affine Transformation.
- (2002). Agilent 10770A Angular Interferometer with Agilent 10771A Angular Reflector,
- An Advanced Angle Metrology System.
- (1993). An introduction to the bootstrap. Monographs on statistics and applied probability.
- (2005). An outline of supplement 1 to the guide to the expression of uncertainty in measurement on numerical methods for the propagation of distributions. Measurement Techniques,
- Analyse des relevés topographiques de l’ESRF,
- (1997). and IEEE Industrial Electronics Society., Capacitive sensors : design and applications.
- Angle Encoders. [WWW] [cited; Technical documentation pertaining to the RON 905 operation principle].
- Angle standards and their calibration,
- (1986). Application of Capacitance Techniques in Sensor Design.
- (1967). Application of the conformal theory of refraction. Zeitschrift fur Vermessungswesen,
- (1998). Applications of x-ray interferometer-generated Moire patterns. Nanotechnology,
- (2002). Applied functional data analysis : methods and case studies. Springer series in statistics.
- (2005). Attempts to include uncorrected bias in the measurement uncertainty. Talanta,
- (2006). Basic Construction of the Flat Angle Calibration Test Bench for Geodetic Instruments. Geodesy and Cartography (Geodezija ir kartografija),
- (2004). Bayesian estimation methods in metrology.
- (2008). calcul incertitude edition 8,
- (2003). Calculating and Converting between Common Vapour Measures
- (2003). Calibration of high-resolution electronic autocollimators against an angle comparator. Metrologia,
- (2006). Calibration of Total Station Instruments at the European Synchrotron Radiation Facility.
- (2006). Calibration of Total Stations Instruments at the ESRF
- (2006). Capacitive Sensor Operation and Optimisation TechNote LT03-0020. Volume,
- Chaîne de mesure dimensionnelle capacitive à sortie linéaire.,
- Classical dynamics of particles and systems.
- Contribution a l'Evaluation de Angles - Conception, Realisation et Validation dún Plateau Pivotant de Tres Haute Precision : Vers une Nouvelle Reference Angulaire Nationale,
- (2005). Correcting capacitive displacement measurements in metrology applications with cylindrical artifacts.
- (1973). Design and Some Applications of Sensitive Capacitance Micrometers.
- (1989). Design Considerations in Multiprobe Roundness Measurement.
- Development of a Dispersometer for the Implementation into HighAccuracy Direction Measurement Systems,
- (2005). Differing angles on angle.
- (1997). Dynamic angle measurement by means of a ring laser. Metrologia,
- (2008). e-Handbook of Statistical Methods,
- (2003). Effect of non-significant proportional bias in the final measurement uncertainty. Analyst,
- Ein interferometrisched Verfahren zur Bestimmung von Strichverbesserungen an eingebauten Theodolitteilkreissn,
- (1991). Electromagnetic distance measurement. 3rd ed ed.
- (1996). Electronic Distance Measurement- An Introduction. 4th ed.
- Electronic Surveying Instruments: A Review of Principles, Problems and Procedures.
- (1988). Electronic Transducers for Industrial Measurement of Low Value Capacitances.
- (1999). European co-operation for Accreditation.
- (2006). Evaluation of measurement data — Supplement 1 to the “Guide to the expression of uncertainty
- (1998). Evaluation of the uncertainty associated with a measurement result not corrected for systematic effects.
- (2006). Evolution of the 'Guide to the Expression of Uncertainty in Measurement'. Metrologia,
- (1993). Four-point method of roundness and spindle error motion measurement. CIRP,
- (1997). Functional data analysis. Springer series in statistics.
- Galileo high-resolution encoder system, in
- (1988). Goniometer with Continuously Rotating Gratings for Use as an Angle Standard. Precision Engineering-Journal of the American Society for Precision Engineering,
- Guide to the Expression of Uncertainty
- (1997). Guidelines for expressing the uncertainty of measurement results containing uncorrected bias.
- (2008). High precision angle calibration of robotic total stations and laser trackers.
- (1994). High-Resolution of Rotary Encoder Analog Quadrature Signals.
- (1976). Improving the Accuracy of Roundness Measurement.
- Inclinomètre à niveaux hydrostatiques de haute résolution en géophysique / High Resolution Water-Tube Tiltmeter in Geophysics,
- Instrumentation and Calibration at the ESRF.
- (2004). Instrumentation and Survey Networks at the ESRF.
- Intercomparaison des méthod de mesur dans le domaine de ls Métrologie Tridimensionnelle par
- (1986). Introduction to space dynamics. Dover books on engineering.
- (2005). La Mesure des Angles au BNM-LNE -Cretion d'une Nouvelle Reference de Mesure Angulaire.
- (2002). Large-scale metrology - An update.
- (2008). Laser Tracker Test Facility at SLAC -
- (1986). Measurement of angle
- Mesure de distance par capteur capacitif et électronique numérique,
- (1999). Mise au Point d'une methode d'étalonnage de théodolites.
- (1997). Modeling and error analysis for assessing spindle radial error motions.
- (2008). Modulo operation
- Multiple view geometry in computer vision. 2nd ed. 2003, Cambridge:
- (2006). Nanometer-level comparison of three spindle error motion separation techniques.
- (2006). Performance Evaluation of Laser-Based Spherical Coordinate Measurement Systems.
- (1999). Practice for non-contacting CMMs :
- Precession and nutation of a gyroscope.
- (2008). Precision Spindle Metrology.
- (1996). Refractive index of air: New equations for the visible and near infrared. Applied Optics,
- Refractive index of air. 2. Group index.
- (1999). Refractive Indices of Light, Infrared and Radio Waves in the Atmosphere,
- (1996). Self Calibration: Reversal, Redundancy, Error Separation,
- (1996). Self Calibration: Reversal, Redundancy, Error Separation, and 'Absolute Testing'. Annals of the CIRP,
- (2005). Self-calibratable rotary encoder,
- (2008). Self-calibration of divided circles on the basis of a prime factor algorithm.
- (2006). Software Specifications for Uncertainty Evaluation,
- (2006). Software Support for Metrology Best Practice Guide
- (2004). Some Reflections on the Validation and Analysis of HLS Data.
- (1976). Some Theoretical Aspects of Error Separation Techniques in Surface Metrology.
- (2008). Spline Toolbox Users Guide, I. The Mathworks, Editor.
- Surveying Instruments.
- (2008). Synchrotron Radiation Facility.
- (2005). Techniques for calibrating spindles with nanometer error motion.
- (2004). The accuracy of angle encoders.
- (2005). The concise Oxford dictionary of mathematics. 3rd ed.
- (2008). The design and application of rotary encoders. Sensor Review,
- (2001). The elements of statistical learning : data mining, inference, and prediction. Springer series in statistics.
- The measure of all things : the seven-year odyssey that transformed the world. 2002, London: Little Brown.
- (2008). The Metre Convention
- (1939). The Refraction and Dispersion of Air for the Visible Spectrum. The Royal Society Philosphical Transactions Mathematical, Physical and Engineering Sciences,
- (1966). The Refractive Index of Air.
- (2005). Treatment of bias in estimating measurement uncertainty. Analyst,
- (2004). Uncertainties in small-angle measurement systems used to calibrate angle artifacts.
- Uncertainty characterization for portable variable geometry coordinate measurement systems.
- Understanding robust and exploratory data analysis.
- (2007). United Kingdom Accreditation Service: Feltham,
- (2002). Use and Calibration of Ultraprecision Axes of Rotation With Nanometer Level Metrology,
- What is a Lock in Amplifier?: Technical Note TN 1000. 2000, Perkin Elmer Instruments: Oak Ridge TN.
- (1961). Zur Reduktion eletronisch gemessener Strecken und beobachteter Winkel wegen Refraktion. Osterreichische Zeitschrift fur Vermessungswesen,

To submit an update or takedown request for this paper, please submit an Update/Correction/Removal Request.